Tohban Report 2014-08-20
From RHESSI Wiki
|Start Date:||13 Aug 2014|
|End Date:||20 Aug 2014|
|Next Tohban:||Lindsay Glesener|
|List all reports|
Solar activity was very low this week.
How many GOES flares occurred?
Flares above B, C, M, X class were 35 9 0 0
And how many of these are listed in the RHESSI flare list?
Flares above B, C, M, X class were 7 8 0 0
And how many had EXCELLENT coverage?
Flares above B, C, M, X class were 5 4 0 0
There were RHESSI flares/GOES flares 94 / 44 over the time range 12-Aug-14 19-Aug-14
Four of the detectors (1, 3, 8, and 9) are stably segmented, with a fifth detector (6) is segmented most of the time, except particularly following SAA passages.
Attenuator logic will be changed from [1, 1, 3, 9] to [1, 1, 3, 3] due to noise on detector 9
2014-226-03:40 UT HV or detector 6 to 4926 V, up from 4500 V
on subsequent two passes: G5 front slow to 0x80 (was 0x70) G6 front slow to 0x50 (was 0x60) G7 front slow to 0x80 (was 0x70)
ACS mode idle 226-15:52 UT, spin rate anomaly 227-00:14, commanded to precession 227-01:40 UT, normal pointing
|HLAT Decimation||Rear decimation weight 6, no front decimation|
|Night time data (fronts)||+-4 minutes|
|Night time data (rears)||+-4 minutes|