Tohban Report 2015-12-16
From RHESSI Wiki
|Start Date:||9 Dec 2015|
|End Date:||16 Dec 2015|
|List all reports|
Decimation was active/vigorous for the full time period. Both front and rear detectors are off during SC night, except for 4 minutes on either side. The SSR is not emptying out.
The cold plates, 1 and 2 were at 136.9 and 135.4 K, on 2015-11-30.
Enhanced fast rates (and reduced livetime) in the rear segment. Because rear-fast threshold is already at 0xFF, we tried lowering HV. No obvious change in the monitor rates.
Dec 14, 18:19, lowered HV from 3600 to 3400 V.
Dec 09, 23:42, raised front fast threshold from 0x70 to 0xFF Dec 14, 18:19, lowered HV from 4100 to 3900 V
Dec 10, 16:42-18:19, power-cycling
Dec 10, 21:38, lowered HV from 1500 V to 1000 V Dec 12, 00:33, events turned on for 3.5 min
No change made
Dec 10, 19:58, lowered front slow threshold from 0x40 to 0x30 Dec 12, 00:30, lowered front slow threshold from 0x30 to 0x20
Dec 10, 21:38, lowered front slow threshold from 0x20 to 0x10 Dec 12, 00:33, lowered HV from 3500 to 3300 V Dec 12, 19:20, raised front slow threshold from 0x10 to 0x20
No change made
|HLAT Decimation||Rear decimation weight 6, no front decimation|
|Night time data (fronts)||+/- 4 minutes|
|Night time data (rears)||+/- 4 minutes|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Attenuator operation||Working as expected|
|Detector problems?||See notes above.|