Tohban Report 2016-05-18
From RHESSI Wiki
|Start Date:||11 May 2016|
|End Date:||18 May 2016|
|List all reports|
The activity has been low throughout the week, mostly at B-level. There are currently 5 ARs on disk and 3 of them are expected to rotate off the disk in the next day or so.
How many GOES flares occurred?
Flares above B, C, M, X class were 25 11 0 0
And how many of these are listed in the RHESSI flare list?
Flares above B, C, M, X class were 11 6 0 0
And how many had EXCELLENT coverage?
Flares above B, C, M, X class were 10 3 0 0
There were RHESSI flares/GOES flares 69 / 36 over the time range 09-May-16 16-May-16
There are issues with the process creating the RHESSI flare list for this week, and these issues will persist until the detector situation is stable, so the RHESSI flare count is an overestimate.
Decimation was active/vigorous for the full time period. The SSR is empty after each day's passes, and is at 0.4% currently (2016-05-17 22:40).
Detectors 1, 3, 8 and 9 are powered on, and are recording events. The cold plates, 1 and 2 are at 136.0 and 134.7 K, respectively (They were 134.6 and 133.2 K last week).
No significant (>300 sec) data gap.
Detectors 1, 3 and 8 are powered on, and are recording events. No excess large numbers of resets or fast or slow rates at present. Detectors 4, 5, 6, 7 were powered up during the week, but did not segment at 5000V, so were turned off.
Attenuator State HEX value deadtime Min A0 0x00 0.00 % Max A0 0x10 6.27 % Min A1 0x07 2.74 % Max A1 0x15 8.24 % Min A2 0x05 1.96 % Max A2 0x29 16.08 % Min A3 0x09 3.53 % Max A3 0xFF 100.00 %
|HLAT Decimation||Rear decimation weight 6, no front decimation|
|Night time data (fronts)||+/- 4 minutes|
|Night time data (rears)||+/- 4 minutes|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Attenuator operation||Working as expected|
|Detector problems?||See notes above.|