Tohban Report 2016-07-20
From RHESSI Wiki
|Start Date:||13 July 2016|
|End Date:||20 July 2016|
|Tohban:||Juan C. Martinez Oliveros|
|Next Tohban:||Juan Camilo Buitrago Casas|
|List all reports|
Solar activity was very low during the week. Currently the activity level is about B3 and no significant solar activity is predicted in the following days. Forecasts show a moderate change for an M-class flare.
How many GOES flares occurred?
Flares above B, C, M, X class were 17 19 0 0
RHESSI flare list contains 6 over the time range 13-Jul-16 20-Jul-16
No overlapping RHESSI flares yet to count, sorry
The SSR has been emptying at the end of the passes set.
Detectors 3 and 8 are active and taking data. The cold plates 1 and 2 temperatures are 133 and 135
The Cryocooler power was increased to ~74.4W on July 14th
No data gaps this week
During this week detectors 3 and 8 were turn on.
16-201-09:54:37 start idib_on_g3 16-201-09:54:53 start idib_on_g8 16-201-09:59:55 start ihv_ramp_stage1_g3_g8 16-201-11:27:18 start ihv_ramp_stage2_g3_g8 16-201-13:06:58 start ihv_ramp_stage3_g3_g8 16-201-13:13:13 /itmon value=3
16-202-09:20:59 start ihv_ramp_stage4_g3_g8 16-202-10:59:43 start ihv_ramp_stage5_g3_g8 16-202-11:03:09 start ihv_ramp_stage6_g3_g8
Also, the attenuator logic using only detector 3 was activated.
|HLAT Decimation||No Rear decimation, no front decimation|
|Night time data (fronts)||full-time|
|Night time data (rears)||full-time|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Attenuator operation||Working as expected|
|Detector problems?||See notes above.|