Tohban Report 2017-07-19
From RHESSI Wiki
|Start Date:||12 July 2017|
|End Date:||19 July 2017|
|Next Tohban:||Mitsuo Oka|
|List all reports|
The Sun was fairly active and region 12665 produced a series of flares. In particular, 2017-07-14, a long duration M-class flare has been observed where RHESSI covered most of the impulsive phase including the peak.
TIME RANGE:12-Jul-17 00:00:00.000 -- 19-Jul-17 00:00:00.000 Flares above B, C, M, X class were 31 24 1 0
How many GOES flares occurred?
Flares above B, C, M, X class were 31 24 1 0
And how many of these are listed in the RHESSI flare list?
Flares above B, C, M, X class were 16 11 1 0
And how many had EXCELLENT coverage?
Flares above B, C, M, X class were 12 4 0 0
There were RHESSI flares/GOES flares 120 / 56 over the time range 12-Jul-17 19-Jul-17
Decimation was active/vigorous for the full time period, with the SSR level currently at around 30%.
Detectors 1, 3, 6, and 8 are recording events. The cold tips are about 128K as of July 19, and cold plates are at about 153K. Cryocooler power is steady at 75.6W, for recent tmperatures, cryocooler efficiencies, and accelerometer values see:
The longer daylight of the current orbits caused the batteries to be on the limit of overcharging. Ops is observing if the battery pressure is crossing the 7000kPa limit and whether the temperatures are spiking towards the end of the daylight portion of the orbit.
Detectors 1, 3, 6, and 8 are powered on, and are recording events. No detector operations this week.
Due to the increase sun fraction, the detector temperature came down. This also caused a lower deadtime for detector 3 which allowed a lowering of the attenuator thresholds.
Det. 1R reset are currently at about 40'000.
With the decreasing sun fraction, the temperatures will likely rise again and the future Tohban will have to watch Det. 3F livetime and Det. 1R reset rates.
On 2017-194-19:50:11 UTC (July 13, 2017), the attenuator thresholds were changed, to reflect a new background of 17.1% deadtime.
0.00 Min A0 22.35 Max A0 <- Thin in 18.82 Min A1 <- Thin out 24.31 Max A1 <- Thick in 18.04 Min A2 32.16 Max A2 19.61 Min A3 <- Thick out
100.00 Max A3
|HLAT Decimation||Rear decimation|
|Night time data (fronts)||plus/minus 4 minutes from daylight|
|Night time data (rears)||plus/minus 4 minutes from daylight|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Attenuator operation||See Notes above|
|Ref. deadtime for Att.thresholds||17.1%|
|Detector problems?||See above|