Tohban Report 2018-05-09
From RHESSI Wiki
|Start Date:||02 May 2018|
|End Date:||09 May 2018|
|List all reports|
Very quiet... GOES baseline level was ~A3, increased a little bit (A5) with the appearance of an active region over the eastern limb.
SSR fill has been slowly creeping up (~2% per day). Currently at 71%.
for a summary plot of recent temperatures, efficiencies and accelerometer values.
Due to high SSR fill, there is no downlinked data for the past three days. No VC-3 data prior to that (normal, detectors are off), but we are getting VC1-SOH and VC-1 (PMTRAS).
Detectors 3 and 8 are powered on, and are recording events. No threshold changes or HV changes this past week.
|Decimation||ANNEALING (previously: Active/Vigorous)|
|HLAT Decimation||Rear decimation|
|Night time data (fronts)||ANNEALING (previously: plus/minus 4 minutes from daylight)|
|Night time data (rears)||ANNEALING (previously: plus/minus 4 minutes from daylight)|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Attenuator operation||No - ANNEALING (previously: Nominal)|
|Ref. deadtime for Att.thresholds||22.x%|