Tohban Report 2018-06-13
From RHESSI Wiki
|Start Date:||6 June 2018|
|End Date:||13 June 2018|
|Tohban:||Juan Carlos Martinez Oliveros|
|List all reports|
Solar activity was very low during the week.
How many GOES flares occurred?
Flares above B, C, M, X class were 1 1 0 0
RHESSI flare list contains 0 over the time range 06-Jun-18 13-Jun-18
No overlapping RHESSI flares to count, sorry
The SSR is probably full by today, because of the anomaly.
All detectors are offline
No info--- on Temperatures
The cold tip 1 and 2 temperatures are ~171.8 and 170.1 K.
Currently the Cryocooler power is ~52 W and the acceleration is at ~21mg.
= Detector issues
During this week the cryocooler power was increased from 46W to 52W. Soon after the power increase, communications with the spacecraft became difficult resulting in the total loss of commanding capability. The ATS ran out at 161-23:23:59 UTC, which means that RHESSI will no longer turn on the transmitter, for scheduled contacts and blind acquisition won't work without an uplink. After the spacecraft goes four days without receiving a command, it will run an RTS which will power off the cryocooler. At that point, the receiver should cool enough to get some commands in.
|HLAT Decimation||Rear decimation|
|Night time data (fronts)||plus/minus 4 minutes from daylight|
|Night time data (rears)||plus/minus 4 minutes from daylight|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Detector problems?||See notes above.|