Tohban Report 2009-07-29
From RHESSI Wiki
|Start Date:||22 July 2009|
|End Date:||29 July 2009|
|Next Tohban:||Meriem Alaoui|
|List all reports|
Quiet. No active regions, no flares.
As of July 24th, 1523 UT, front segments are now being taken during eclipse. The SSR is still only maxing at 25% so this seems ok. (SSR min is 0%.)
Due to bad timing in the automated system, the monitor rates were left enabled for two orbits this morning. That segment of the memory filled up and so we lost ~25 minutes of data.
No other gaps.
G8 was lowered by 4 steps and remains segmented.
The threshold on detector 8 will be adjusted
A discussion regarding whether or not to anneal is tentatively scheduled for August 12.