Tohban Report 2012-08-01
From RHESSI Wiki
|Start Date:||25 July 2012|
|End Date:||01 August 2012|
|Next Tohban:||Nicole Duncan|
|List all reports|
Solar activity was moderate this week, with 12 active regions currently on the disk. There were few large flares, but base GOES level is high, with many C flares. Since we are currently in active/vigorous decimation and the SSR is emptying each day, we are in a good position to handle data if there should be a large flare coming up. It is expected that activity will remain at its current level in the next few days. (STEREO-B does not show many new regions coming, but some of the current regions have just barely turned earthside.)
How many GOES flares occurred?
Flares above B, C, M, X class were 8 43 4 0
And how many of these are listed in the RHESSI flare list?
Flares above B, C, M, X class were 6 29 1 0
And how many had EXCELLENT coverage?
Flares above B, C, M, X class were 5 16 0 0
There were RHESSI flares/GOES flares 300 / 55 over the time range 18-Jul-12 25-Jul-12
The SSR is emptying after each pass set.
There was a 10-minute gap of lost data in monitor rates around 2:30 on the 27th.
Last week the HV on detector 8 was lowered in order to reduce noise coming out of the SAA. After the HV change, detector 8 continued to show excessive count rates but not so high as they were previously. Since the livetime is not greatly affected by this and the memory situation is ok, no further action was taken for this detector. Detector 2 is also occasionally displaying high rates due to the ongoing light leak problem. Since the percentage of observing time that is affected is not great, and raising the threshold would impair low-energy flare spectroscopy, no action was taken on this problem either.
Brian has prepared further long-term plots of the cold plate temperature rise issue. Discussion on this topic was postponed until a time when Stuart and Rob can call in (possibly next meeting?)
AGU deadlines are coming up! Abstract deadline is next Wednesday; membership renewal should be done by this Friday.
|HLAT Decimation||Rear decimation weight 6, no front decimation|
|Night time data (fronts)||+/- 4 minutes|
|Night time data (rears)||+/- 4 minutes|
|Require extra passes?||No|
|Requirement for moving pointer?||No|
|Detector problems?||Detectors 8 and 2 noisy; no actions taken.|